Cryo Mini Hose - Extended
- Zimmer MedizinSystems
Zimmer MedizinSystems Cryo Mini Extended Hose
Cryo Mini ExtendedHose
Replacement extended hose for Zimmer MedizinSystems Cryo Mini models
Length: 12 feet
Essential part for cold air delivery
Premium quality, durable material
Works with Zimmer MedizinSystems Cryo Mini articulating arm and focusing tip
Cryo Mini Extended Hose - Increased Maneuverability
The Zimmer MedizinSystems Cryo Mini extended hose is intended for use with the Zimmer MedizinSystems Cryo Mini skin cooling system. Only Zimmer MedizinSystems Cryo Mini models should be equipped with the Cryo Mini extended hose. Other Zimmer MedizinSystems Cryo models should be fitted with their corresponding Cryo hoses and/or extended hoses.
The Zimmer MedizinSystems Cryo Mini extended hose delivers cold air of up to -10 C from the Zimmer MedizinSystems Cryo Mini skin cooling system to the treatment site. The cold air is typically intended to minimize pain and thermal injury during laser and dermatological treatments and for temporary topical anesthetic relief for injections. The epidermis is cooled before, during, and after laser energy has been applied – without interfering with the laser beam.
The Cryo Mini extended hose is 12 feet in length and made of premium quality, durable materials. The Zimmer MedizinSystems Cryo Mini extended hose is especially suited for larger treatment rooms or treatment rooms that host several pieces of equipment where moving the Cryo Mini unit is not practical. The Cryo Mini extended hose can also be of benefit if placed between multiple treatment chairs. The Cryo Mini extended hose attaches to the front of the Cryo Mini unit. The extended hose is installed by lining up the pin on the extended hose with the indentation on the air outlet of the Cryo Mini unit. Then turn the extended hose clockwise and pull slightly down to secure in place. To release the Cryo Mini extended hose from the Zimmer MedizinSystems Cryo Mini unit, simply push the extended hose slightly up, turn counterclockwise, and the extended hose will slide out.
Replacement of the Cryo Mini extended hose is recommended when you see or feel air leaks or compression points. With the Cryo Mini unit in operation run your hand across the entire length of the extended hose looking and feeling for air leaks. Using a damaged extended hose can restrict air flow and affect the air load delivered. Please do not pull or stretch the extended hose beyond its usable distance as overextending the extended hose can cause damage and result inefficient treatment and/or damage to the Zimmer MedizinSystems Cryo Mini unit. Rather wheel the Cryo Mini unit to a more suitable location before beginning treatment.
Works with the following accessories:
Zimmer MedizinSystems Cryo Mini articulating arm
The Zimmer MedizinSystems Cryo Mini articulating arm allows the user hands-free operation of the Cryo Mini unit. The Zimmer MedizinSystems Cryo Mini articulating arm holds the extended hose in place during treatment and can be repositioned as needed. Sold separately.
Zimmer MedizinSystems Cryo Mini focusing tip
The Zimmer MedizinSystems Cryo Mini focusing tip is an air nozzle that provides the user with a more concentrated air flow. The focusing tip attaches to the extended hose by inserting the tip into the treatment end of the extended hose. The range of the tip can be adjusted by sliding it more or less into the Cryo Mini extended hose, and to provide a means to reach those harder to reach treatment sites. Sold separately.